Throughout this manual, the following conventions are used for SCPI command syntax. • Square brackets .. HEWLETT-PACKARD,EA,0,X.X-X.X-X.X. Read about ‘EA Triple Output DC Power Supply User’s Guide’ on element14 .com. EA Triple Output DC Power Supply User’s Guide. Items are supplied with manuals, accessories and typically a full no-quibble 2 . HP EA/32A/33A/34A Programmable DC Power Supply Specifications.
|Published (Last):||16 July 2010|
|PDF File Size:||4.65 Mb|
|ePub File Size:||13.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
A record of up to 20 errors can be stored in the power supply’s error queue. With power-off, connect a load to the desired output terminals.
Contact the seller – opens in a new window or tab and request a shipping method to your location. For example, the following statements executed from your computer will read the power supply’s error queue and print the most recent error: Contact the seller – opens in a new window or tab and request a shipping method to your location.
Sandwich, Illinois, United States. Mouse over to Zoom – Click to enlarge. Subject to credit approval. The spiral spline is tight and intact with NO splits. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab. Orders are not processed for shipping on weekends or holidays.
Instrument Control (GPIB, Serial, VISA, IVI)
The item may have some signs of cosmetic wear, but is fully operational and functions as intended. I realize that when using the power supply in manual operation, as well as in remote operation with specific drivers I have to enable all outputs at once, I cannot enable each output separately. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, manuaal, brokerage and other fees.
Message 1 of Minimum monthly payments are required.
HP EA Triple Output DC Power Supply Service Manual E B-3 | eBay
Select a valid country. Page 8 Tutorial Chapter 7 describes basic operation of linear power supplies and gives specific details on the operation and use of the Agilent EA power supplies.
Most Active Hardware Boards: You will observe the following sequence on the display after pressing the front panel power switch to 1 All segments of the display including all annunciators will turn on for about one second. Page 34 Chapter 2 Initial Operation Output Checkout 6 Ensure that the current can be adjusted from zero to the maximum rated value.
There are 1 items available. Output Checkout Chapter 2 Initial Operation Output Checkout Output Checkout The following procedures check to ensure that the power supply develops its rated outputs and properly responds to operation from the front panel. Page 5 N o t e All front panel keys and controls can be disabled with remote interface commands.
Hewlett-Packard HP EA Triple Output D.C. Power Supply User’s Guide Manual | eBay
See the Service Guide for instructions on returning the power supply to Agilent Technologies for service. General Information This chapter provides a general description of your power supply.
Before installation or operation, check the power supply and review this manual for safety markings and instructions.
Refer to eBay Return policy for more mankal. Message 10 of Sell now – Have one to sell? For Safety information, Warranties, and Regulatory information.
E3631A 80W Triple Output Power Supply, 6V, 5A & ±25V, 1A
Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc. The Agilent EA contains three linear regulated power supplies. Page Tutorial The Agilent EA is a high performance instrument capable of delivering clean dc power. But to take full advantage of the performance characteristics designed into the power supply, certain basic precautions must be observed when connecting it for use on the lab bench or as a controlled power supply.
Although the APPLy command provides the most straightforward method to program the power supply, e3613a low-level commands give you more flexibility to change individual parameters. An external source can not be allowed to pump current into the supply without risking loss of regulation and possible damage. This item will ship to United Statesbut the seller has not specified shipping options.
Always refer to the English version at the Agilent Web site for the latest revision. Any international shipping and import charges are paid in part to Pitney Bowes Inc. For example if one of the three supplies is in constant voltage mode and due to an overload loses regulation, bit 13 is set latched. Minimum monthly payments are required. This chapter also contains instructions for initial inspection, location and cooling for bench and rack operation, selecting the power-line voltage, and connecting your power supply to ac power.